Compact size and light weight, convenient for patients to carry
Cordless and hand-held, flexible to use
Easy operation, suitable for home use
With microcomputer timer function
The LCD screen direction can be adjusted if needed
Use special optical materials acrylic shield to provide a more secure protection
Unique comb for distance guide as well as the scalp treatment
Steps of Operation:
A. Confirmation of radiation dose: According to the radiation intensity of the phototherapy instrument under the doctor's recommendation.
Example: The radiation intensity of the phototherapy instrument is E=1mw/cm2, the doctor recommends that each dose is Ψ=0.5J, and the required time is T=8 minutes and 20 seconds;
B. Connect the ballast to the power supply;
C. Check the clock, write down the start time, and calculate the end time;
D. Place the irradiator above the affected area and turn on the power switch;
E. When the time is over, turn off the switch and disconnect the power supply.
Precautions for use:
1. The irradiation distance has a great influence on the irradiation intensity, and the correctness of the irradiation distance should be ensured;
2. After phototherapy, the patient finds continuous lumps, pain or pigment spots on the skin, and should consult a dermatologist and take corresponding measures.
1. Please use it under the guidance of a doctor, do not use it without authorization.
2. The instrument should be placed in a safe place that is not easy for children to reach, to prevent children from mistaking it and hurting eyes and other parts.
Item No.: YK-6000B
Size: 275*55*20mm (mm)
Weight: 1.115 kg
Working distance: 3CM
Rated power: 20VA
Output waveform type: UVB
Effective irradiation area: 48cm²±2%
Working voltage: 110V/60HZ
UVB Lamp parameters:
Size: 28×165 mm (H type)
Average life: 2000 hours
Irradiation intensity: 3-16mw/cm²
Spectral range: 280-320mm