VRN-Q5 No-pain ultrasonic Piero treatment combines scaling,periodontal and endodontic on one device ,they can meet the most needs of supragingival and subgingival scaling operations:
* The scaler tip is casted from Titanium alloy to ensure circular vibration track and more concentrated amplitude, so that Period, Scaling. Endo, Restorative treatments can be done simultaneously.
* Two large refillable bottles , satisfy using pure water or disinfectant.
* The hydraulic routing is made from antiseptic material. Disinfectant for the treatment, such as hydrogen peroxide, chlorhexidine, sodium hypochlorite can be filled into the automatic water supply system.
* Detachable hand piece can be 135℃and 0.22 MPA autoclave sterilization.
* The evolutionally wireless foot pedal remotes the main unit
* Soft bright LED light improves the clinical efficiency.
Power input: AC110V/60HZ
Input power: 20-40 W
Wireless foot switch battery: AA battery × 2
Receiving sensitivity -114 dB (in accordance with China's National Telecommunications Law) Receiving frequency: 2.4 G-2.5 G
The main vibration deviation of the tip of the tip:
Minimum value, 1μm, deviation -50%,
Maximum value, 100μm, deviation +50%
Half offset force:
Minimum value, 0.1 N, deviation -50%
Maximum value, 2 N, deviation +50%
Tip vibration frequency: 25 kHz～35 kHz;
Note: The vibration frequency of the tip of the working tip of different models is different, but they are all distributed in the range described.
Cutting-edge output power: 3 W～20 W
Fuse: T1AL 250V
Host weight: 1.8 kg
Operation mode: continuous operation
Type of protection against electric shock: Class Ⅱ equipment
Degree of protection against electric shock: BF type application part
Degree of protection against ingress of liquid: ordinary equipment (IPX0), wired foot switch is waterproof equipment (IPX1), wireless foot switch is waterproof equipment (IPX4)
The degree of safety when used in the case of flammable anesthetic gas mixed with air or flammable anesthetic gas mixed with oxygen or nitrous oxide: non-AP, APG-type equipment
Wireless foot switch: transmit frequency 2.412 GHz-2.462 GHz, modulation type GFSK, effective radiation power 12dBm
a) Ambient temperature: 5℃～40℃
b) Relative humidity: ≤80%
c) Atmospheric pressure: 70kPa～106kPa
d) Applicable range of power supply voltage: 220 V±22 V～
Disinfection in high temperature and high pressure environment
1) When using a sterilizer for disinfection, 135℃, 2.2bar (0.22MPa), 15 minutes.
2) Pull out the handle and remove the working tip after each use.
3) Wrap the handle with a sterile towel or bag.
4) After disinfection, the handle should be cooled naturally before it can be used again to avoid burns.
5) When this product is not used for a long time, it should be connected to water and electricity once a month, 5 minutes each time.
This product should be stored in an environment with a temperature of -20℃～55℃, a relative humidity of 10%～90%, and an atmospheric pressure of 70kPa～106kPa
Main Unit × 1
PH-2 LED ultrasonic Piezo Handpiece × 2
Wireless Foot Pedal × 1
Water Bottle × 2
Power Supply Cord × 1
Perio Tips × 10
Implant Tip × 1
Scaler Tips ×4
Endo Tip × 1
20# Endo File × 6
25# Endo File × 6
Disinfection Box × 2
Torque Tip Wrench × 1
Endo File Wrench × 1
Wired Foot Pedal × 1